MITECH MHBS-62.5 Digital Display Small Load Brinel
Mitech MHB-62.5 Digital Display Small Load Brinell Hardness Tester, based on the mechanical principle of hard alloy indenter pressing into the sample surface to produce indentation, realizing the material hardness measurement by measuring the diameter of the indentation. It’s MITECH latest optical mechanical and electrical integration of high-tech products. It is novel and high reliable with accurately measurement. With LED display, simple operation, it can visually display the test results to meet the hardness testing requirement for the quality control and qualified assessment of the workpiece sample. It is widely used in metal processing and manufacturing, various metal material’s failure analysis and other fields like colleges and research institutions.It is the new type Brinell hardness testing instrument for testing the hardness of the materials like cast iron, steel, soft alloy with small and thin sheet type workpiece and so on.
Function & application
l Metal processing industry quality control links
l Universities teaching and demonstration test
l The failure test of metal material
l The material hardness test of scientific research institutions
Hardness is not a simple physical quantity, but a reflection of the material elasticity, plasticity, strength and toughness .and hardness test is the most simple mechanical testing method to determine the metal material performance. Also one of the important means to judge the products quality.
Brinell hardness test: Test force(F) is on the steel ball with certain diameter(D) , pressed on sample surface. After a period of time, cancel the force. The indentation diameter is get by measuring with micrometer ocular,thus to calculate the average pressure(N/mm²).
l Working Temperature：18℃～28℃；
l Relative Humidity：≤65％；
l Clean environment, no vibration；
l No corrosive media around.
l Suitable for measuring the small or thin workpiece hardness , a variety of small load Brinell scale selection;
l Support Brinell, Rockwell, Vickers and other hardness standard conversion
l Built-in large-capacity chip, can store large amounts of data, comes with the database for comparison;
l Modeling novel, strong structure, high reliability, simple operation, intuitive readings, high test efficiency;
l With RS-232 interface, can communicate with the computer;
l Equipped with excellent performance of the carbide indenter, high hardness, wear resistance, good toughness, while high
corrosion resistance, to ensure that the instrument test value is accurate, stable and reliable;
l Host stability is good, the workpiece surface quality and man-made factors on the hardness of the test results less impact;
l LCD display, menu smart tips, easy to operate;
l Optional shooting device, can achieve the measured indentation and material metallographic organization to shoot;
l Compliance with ISO 6506, ASTM E10-12, ASTM E-384, GB / T231.2, JIS Z2243 and other relevant domestic and foreign
Operating Method and Attentions
l Before using this instrument, please read the instruction manual carefully, learn more about the steps and precautions.
l In general, the first measurement point is only used to offset the gap between the support surface, without recording the
test results, from the second point of statistical test results;
l Machine add unloading test power, it is forbidden to turn the variable load the handwheel, such as rotating dislocation
can lead to internal gear, chaos test force, and even cause damage to items within the machine;
l User’s should follow the operating procedures, before and after the test should be repeated with standard block proofing
instrument. If a long time idle, re-boot after several proofreading, to be stable after the test results, and then test part;
l After applying the main test force in the hardness test, it is forbidden to turn the variable hand wheel.
l The instrument should be carried at the bottom of the handling, to prohibit cross. Before loading, please unplug the
power plug, fixed measuring lever and remove the weight and boom;
l When not in use for a long time, please cover the dust cover to prevent dust into the instrument interior. And
the hardness block, indenter should be coated with antirust oil to prevent rusting after using.
l Start using again after long time idling, please make testing after preheating the electrical components.
l The accuracy of the reading microscope has been adjusted before leaving factory. It is prohibited to disassembly without
authorization. Otherwise, it will reduce the reading accuracy of the micro eyepiece.
l The storage and using of the reading microscope should avoid dust, damp and corrosive gas environment;
l If there is dirt on the surface of the reading microscope lens, please wipe it with soft absorbent cotton or lens paper. If
there is oil pollution, please wipe it slightly by the absorbent cotton with industrial alcohol or ether.
l Should be regularly in the screw and hand wheel contact surface into a small amount of oil;
l Conducted at least once a year instrument verification, to ensure the accuracy of the hardness tester;
l Don’t disassemble the instrument without authorization, maintenance related matter, please contact MITECH after-sale service department with 4000600280.
The power series 61.25N（6.25kgf）、98.07N（10kgf）、153.2N（15.625kgf）、 294N（30kgf）、306.25N（31.25kgf）、612.9N（62.5kgf） Measuring range 8~650HBW、8~450HBS Indication accuracy ±3%（HBW≤225） Brinell scale HBW2.5/6.25、HBW1/10、HBW2.5/15.625、HBW1/30、 HBW2.5/31.25、HBW2.5/62.5、HBW5/62.5 Testing Force Application Mode Automatic (loading, holding, unloading) Indenter objective lens conversion mode Manual operation display usage LCD Liquid crystal display Microocular magnification 40X 100X Test force holding time 5~60s Minimum division 0.1μm Applicable material maximum height 160mm Max distance from head to body 135mm Voltage AC 220V/50Hz Size 540* 220*650mm Total Weight 45kg
Please contact online service for manual.