Mitech MHRS-150-P Digital Plastic Rockwell Hardnes
Mitech MHRS-150-P Digital Plastic Rockwell Hardness Tester, based on the mechanical principle of hard alloy indenter pressing into the sample surface to produce indentation, realizing the material hardness measurement by measuring the depth of the indentation. Capable of inspecting the finished or semi-finished parts of the machined sample, it is suitable for plastic, hard rubber, synthetic resin, friction materials and softer metals. According to statistics, Rockwell hardness testing is the most widely used hardness testing method in metal processing industry, which utilization ratio is more than 70%. With novel appearance, stable performance, intuitive and convenient LCD display, and easy to operate the menu design, it is widely used in the quality control of plastic products, the detection of finished products of resin rubber and other fields like colleges and research institutions. it is the the sophisticated detection equipment to test the plastic and soft metal materials Rockwell hardness.
Function & application
l Used for quality control in plastic products;
l Used for resin rubber finished product testing
l Demonstration experiment for education and teaching in Colleges and Universities;
l Hardness testing of materials in scientific research institutions
The Rockwell hardness test is taking the diamond cone with 120°apex angle or the hardened steel ball with specified diameter as the indenter to press into sample surface with specific test force, then get the Rockwell hardness of the measured metallic materials according to the sample surface indentation depth.
The Rockwell hardness measurement principle is shown as below figure. 0-0 is the position that the diamond indenter is not yet in contact with the sample. 1-1 figure is the indenter position under the affect of the preliminary test force, the indentation depth is h1. The preliminary test is to eliminate the influence to the testing result accuracy caused by the roughness of the sample surface. 2-2 in the figure is the indenter position under the influence of the testing force (the preliminary test force and the main test force). The depth is h2. 3-3 in the figure is the indenter position after dismounting the main test force. As the metal elasticity will recovery some degree after deformation, the really indentation depth of the indenter is h3. The plastic deformation caused by the main test force make the indenter pressing into the depth is h＝h3－h1. Rockwell hardness value is determined by the size of h, the greater the depth h, the lower the hardness, otherwise, the higher the hardness.
In the traditional concept, usually use a constant C minus h to represent the level of hardness, while the depth of indentation per 0.002mm as a unit of hardness. The hardness value obtained is called the Rockwell hardness value, denoted by the symbol HR.
In the formula, c is a constant (for HRC, HRA, c is 0.2; for HRB, c is 0.26). The Rockwell hardness value HR obtained is an unknown number which is usually read directly on the test machine indicator when testing.
It should be noted that the measured hardness values would be different with different indenter and test force. Therefore, the Rockwell hardness testing specifies 15 different hardness test scales according to the different indenter specification and test force sizes. And the HRB, HRC, HRA are the most widely used.
l Operation Temperature：10～30℃；
l The surrounding environment should avoid of vibration, strong magnetic field, corrosive medium and heavy dust.
l Widely used for high-precision hardness testing for plastic, hard rubber, synthetic resin, friction material and softer metal;
l Adopt electronic automatic operation system to control the testing force, easy to operate;
l Option for various specifications of the indenter, support many types of Rockwell hardness scales testing ;
l Equipped with high-speed thermal printer, quickly print out the test data;
l Support the conversion among various hardness scales such as Brinell, Vickers and etc;
l Adopt large-screen LCD, easy to operate, visually display the test results;
l Equipped with excellent performance of the carbide indenter, high hardness, wear resistance, good toughness,
meanwhile with high temperature and corrosion resistance to ensure that the instrument test value is accurate, stable and
l With the function of threshold overrun automatic alarm, apply to the bulk of finished products or semi-finished pieces of
l The function of original ambient temperature real-time display can avoid the instrument working in the case of high or low
temperature for a long time , resulting in increased test error and reducing the service life;
l Consistent with GB7407、JJG884、ASTMD785、GB/T 3398.2 other relevant standards at home and abroad.
Operating Method and Attentions
l The test personnel should follow the operating procedures.Before and after the test,it should be repeated with standard
block calibration equipment;
l If the specimen has visible deformation marks on the back and edges, the test result would be useless. It should choose
the intender or test force to test again;
l Normally, the first point is only used to offset the gap of each support surface without needing to record the test result.
Record the statistical test result from the second point;
l During testing, to ensure accuracy, please make sure the distance between the adjacent indentation center and the
distance from the indentation center to edge of the specimen is greater than 3mm;
l Pay attention to palm the instrument for carrying to forbid falling across. Please fix well the measuring lever, counterweight
and boom before carrying;
l It is strictly prohibited to turn the variable loading handwheel when the instrument load or unload the test force. Rotating
the handwheel forcibly will cause the internal gear dislocation, test force in chaos and even damage the internal parts of
l Turn off the power after using the instrument;
l When not in use for a long time, please cover the dust cover to prevent dust into the instrument interior. And the
hardness block, indenter should be coated with antirust oil to prevent rusting after using;
l Long time after the re-enabled,it should be preheated the electrical components before test
l The hardness tester should be calibrated at least once a year to ensure its accuracy;
l Don’t disassemble the instrument without authorization, for maintenance related matters, please contact MITECH
after-sale service department with 4000600280.
Preliminary testing force 98.07N，tolerance±2.0% Testing force 588.4N，980.7N，1471N，tolerance±1.0% Measuring range HRA：20-88、HRB：20-100、HRC：20-70、HRD：40-77、 HRE：70-100、HRF：60-100、HRG：30-94、HRH：80-100、 HRK：40-100、HRL Testing force application mode Automatic operation (preliminary test needs manual operation) Indenter specification Φ3.175mm steel ball indenter；Φ6.35mm steel ball indenter；Φ12.7mm steel ball indenter Display LCD Rockwell scale HRA、HRB、HRC、HRD、HRE、HRF、HRG、HRH、HRK Conversion scale HRA、HRB、HRC、HRD、HR15N、HR30N、HR45N、 HR15T、HR30T、HR45T、HV、HK、HBW Duration time 1~30s(The testing force duration time for plastic Rockwell hardness test is 15 seconds) Indication error 0.1HR Maximum height of specimen 170mm Distance of indenter to outer wall 165mm Power supply AC220V/50Hz Dimensions 550*220*730mm Main unit weight 85kg
Please contact online service for manual.